Isolated Organ and Tissue 2
This Heated Water Bath is for the most demanding research work.
It has two heaters having 250 watts of combined power.
The temperatur is automatically controlled by a sensitive thermistor
in a metal tube inside the bat to hold the bath water
temperature has stabilized. A magnetic stirrer in the bottom of the
tank slowly circulates the water to ensue uniform temperature.
These heaters are mounted through the front of the tank near the
bottom and extend 100 mm (4 in) into it. The desired temperature of
the Water Bath is set by a dial on the front panel that is graduated
from 15o to 45o C in 5o increments.
Two on/off switches and indicator lamps, one for power and the other
for the magnetic stirrer are on the front panel.
Download:Isolated Organ and Tissue 2